This week I tested my Palomar QYAG5 against the Hoya Conbio Revlite laser. I planned to demo the Medlite C6, which is almost the same unit, but the Revlite is Conbio’s top of the line machine and of course they want to show the more expensive unit. The touch screen that the Revlite offers over the Medlite C6 is very cool, but when it comes right down to it, the Medlite C6 is really just as good for tattoo removal.
Before I get into the full results, I was honestly shocked how similar the systems really are. I was actually able to produce almost identical results with just one exception that I will address here in a minute.
Here are the results:
Power – The Revlite is a 220V machine, where mine is a 110V. As with the last laser I demoed this machine produces a lot of heat. Also, the Revlite took a while to warm up, whereas the Palomar starts immediately.
Energy – The Revlite does seem to be able to produce more energy, although, the Palomar appears to be able to match it within certain parameters. Looking at the display, the Revlite can produce more power in a larger spot size, but it was more energy than needed to treat the two test clients.
Wavelength – The Revlite has two dye polymer hand pieces that allow the machine to produce 585 and 650 wavelengths that the Palomar cannot. The Palomar has a wave-blending feature that the Revlite does not. Testing light blues the machines have similar results. However, the Revlite does seem to react slightly better to greens.
Spot Size – In 1064 and 532 modes the Revlite does have a 8mm spot size, where the Palomar only has 6mm. When it comes to doing very big black or red pieces, the Revlite can do them a bit faster. However, in 585 and 650 mode the Revlite spot size drops down to 2mm to 3mm making progress super slow going.
Speed – First, as mentioned above, the Revlite needs to warm up, where the Palomar is ready to operate when it’s turned on. Simple, but clients don’t like to wait. As for the operation, in 1064 and 532 modes, both machines operate at the same speed, 10HZ, they fire 10 shots a second. When the Revlite is in 585 and 650 mode it drops down to 1 or 2 HZ making treatment of large blue/green tattoos incredibly time consuming.
Ergonomic – The Revlite articulated arm is slightly more comfortable to use than the larger Palomar lasing unit. The Revlite might be easier to use for some people, but the Palomar also comes with an arm to hold the large hand piece. My thoughts are that the Revlite is more ergonomic, but only slightly.
Function – Side by side these machines are so similar. The only complaints I got from the clients were that the Revlite is just too slow in blue and green modes. When the Palomar operates in the wave blending mode to treat these colors it is so much faster.
Results – I worked on two clients and the results are yet to be seen, give me a week or two. I can tell you from first glance, the results seem almost identical.
Pain – Because the machines operate in the same way, pain seems similar as well. Both clients gave almost identical reviews about the sensation.
Hair Removal – Another reason I wanted to review this machine is its hair removal feature. Not that I plan to delve into this business, but if the machine has the option I might offer it as well. As it turns out, this laser really isn’t designed to do hair removal. During the machine’s initial testing, they found less hair growth in areas that were treated for tattoo removal. So, does the machine do hair removal? Well a bit, but then again, the Palomar does the exact same thing.
In final review, the machines look and operate very differently, but the core physics are extremely similar. Prior to testing the machine I was convinced the Hoya Conbio Revlite and Medlite C6 would have so much more to offer than the Palomar QYAG5. The final test will be in the results, but for now, I see such little difference that I doubt I will be in any hurry to change lasers.
Coming soon, a review of a new pain reduction device.